Full Text View
Aerosol generation related to respiratory interventions and the effectiveness of a personal ventilation hood
Forbes McGain, Ruhi S Humphries, Jung Hoon Lee, Robyn Schofield, Craig French, Melita D Keywood, Louis Irving, Kevin Kevin, Jim Patel, Jason Monty
Crit Care Resusc 2020; 22 (3): 212-220
- Forbes McGain 1, 2, 3
- Ruhi S Humphries 4
- Jung Hoon Lee 5
- Robyn Schofield 6
- Craig French 1, 3, 7
- Melita D Keywood 4
- Louis Irving 8
- Kevin Kevin 5
- Jim Patel 9
- Jason Monty 5
A patent has been filed for the personal ventilation hood by the University of Melbourne/Western Health. Forbes McGain and Jason Monty were the leads in this patent application. All other authors have no conflicts of interest.
OBJECTIVE: To quantify aerosol generation from respiratory interventions and the effectiveness of their removal by a personal ventilation hood.
DESIGN AND SETTING: Determination of the aerosol particle generation (in a single, healthy volunteer in a clean room) associated with breathing, speaking, wet coughing, oxygen (O2) 15 L/min via face mask, O2 60 L/min via nasal prongs, bilevel non-invasive positive-pressure ventilation (BiPAP) and nebulisation with O2 10 L/min.
INTERVENTIONS: Aerosol generation was measured with two particle sizer and counter devices, focusing on aerosols 0.5–5 μm (human-generated aerosols), with and without the hood. An increase from baseline of less than 0.3 particles per mL was considered a low level of generation.
MAIN OUTCOME MEASURES: Comparisons of aerosol generation between different respiratory interventions. Effectiveness of aerosol reduction by a personal ventilation hood.
RESULTS: Results for the 0.5–5 μm aerosol range. Quiet breathing and talking demonstrated very low increase in aerosols (< 0.1 particles/mL). Aerosol generation was low for wet coughing (0.1 particles/mL), O2 15 L/min via face mask (0.18 particles/mL), and high flow nasal O2 60 L/min (0.24 particles/mL). Non-invasive ventilation generated moderate aerosols (29.7 particles/mL) and nebulisation very high aerosols (1086 particles/mL); the personal ventilation hood reduced the aerosol counts by 98% to 0.5 particles/mL and 8.9 particles/mL respectively.
CONCLUSIONS: In this human volunteer study, the administration of O2 15 L/min by face mask and 60 L/min nasal therapy did not increase aerosol generation beyond low levels. Non-invasive ventilation caused moderate aerosol generation and nebulisation therapy very high aerosol generation. The personal ventilation hood reduced the aerosol counts by at least 98%.
- World Health Organization. Clinical management of severe acute respiratory infection (SARI) when COVID-19 disease is suspected: interim guidance. WHO, 2020. https://www.who.int/publications-detail/clinical-management-of-severe-acute-respiratory-infection-when-novel-coronavirus-(ncov)-infection-is-suspected (viewed Apr 2020)
- Shiu EY, Leung NH, Cowling BJ. Controversy around airborne versus droplet transmission of respiratory viruses: implication for infection prevention. Curr Opin Infect Dis 2019; 32: 372-9
- Bourouiba L. Turbulent gas clouds and respiratory pathogen emissions: potential implications for reducing transmission of COVID-19. JAMA 2020; doi: 10.1001/jama.2020.4756. [Epub ahead of print]
- Asadi S, Bouvier N, Wexler AS, Ristenpart WD. The coronavirus pandemic and aerosols: does COVID-19 transmit via expiratory particles? Aerosol Sci Technol 2020; 54: 635-8
- Morawska L, Cao J. Airborne transmission of SARS-CoV-2: the world should face the reality. Environ Int 2020; 139: 105730
- Johnson G, Morawska L, Ristovski Z, et al. Modality of human expired aerosol size distributions. J Aerosol Sci 2011; 42: 839-51
- Costello A, Abbas M, Allen A, et al. Managing the health effects of climate change. Lancet 2009; 373: 1693-733
- Tran K, Cimon K, Severn M, et al. Aerosol generating procedures and risk of transmission of acute respiratory infections to healthcare workers: a systematic review. PLoS One 2012; 7: e35797
- Australian and New Zealand Intensive Care Society. COVID-19 guidelines; version 2. Melbourne: ANZICS, 2020. https://www.anzics.com.au/wp-content/uploads/2020/04/ANZI_3367_Guidelines_V2.pdf (viewed Apr 2020)
- Department of Health and Human Services. Design guidelines for hospitals and day procedure centres — Part E: building services and environmental design. Melbourne: State Government of Victoria, 2004. http://www.healthdesign.com.au/vic.dghdp/ (viewed May 2020)
- AusPat. Portable isolation unit (No. 2020901125). Australian Government, Intellectual Property Australia; 2020. http://pericles.ipaustralia.gov.au/ols/auspat/applicationDetails.do?applicationNo=2020901125 (viewed Apr 2020)
- Shen S, Jaques PA, Zhu Y, et al. Evaluation of the SMPS–APS system as a continuous monitor for measuring PM2.5, PM10 and coarse (PM2.5–10) concentrations. Atmos Environ 2002; 36: 3939-50
- Baron PA, Mazumder MK, Cheng, YS. Chapter 17: Direct-reading techniques using particle motion and optical detection. In: Baron PA, Willeke K,YS; editors. Aerosol measurement: principles, techniques, and applications; 2nd ed. New York, USA: Wiley and Sons, 2001
- Flagan RC. Chapter 18: Electrical techniques. Cheng YS. In: Baron PA, Willeke K,YS; editors. Aerosol measurement: principles, techniques, and applications; 2nd ed. New York, USA: Wiley and Sons, 2001
- Chapter 19 Condensation detection and diffusion size separation techniques. In: Baron PA, Willeke K,YS; editors. Aerosol measurement: principles, techniques, and applications; 2nd ed. New York, USA: Wiley and Sons, 2001
- International Organization for Standardization. ISO 14644-1: 2015 — Cleanrooms and associated controlled environments. Part 1: Classification of air cleanliness by particle concentration. https://www.iso.org/standard/53394.html (viewed Apr 2020)
- Loh NW, Tan Y, Taculod J, et al. The impact of high-flow nasal cannula (HFNC) on coughing distance: implications on its use during the novel coronavirus disease outbreak. Can J Anaesth 2020; doi: 10.1007/s12630-020-01634-3. [Epub ahead of print]
- Leung CCH, Joynt GM, Gomersall CD, et al. Comparison of high-flow nasal cannula versus oxygen face mask for environmental bacterial contamination in critically ill pneumonia patients: a randomized controlled crossover trial. J Hosp Infect 2019; 101: 84-7
- Iwashyna TJ, Boehman A, Capelcelatro J, et al. Variation in aerosol production across oxygen delivery devices in spontaneously breathing human subjects [preprint]. medRxiv 2020; doi: 10.1101/2020.04.15.20066688
- Cook T, El‐Boghdadly K, McGuire B, et al. Consensus guidelines for managing the airway in patients with COVID‐19. Anaesthesia 2020; 75: 785-99
- Lindsley WG, Pearce TA, Hudnall JB, et al. Quantity and size distribution of cough-generated aerosol particles produced by influenza patients during and after illness. J Occup Environ Hyg 2012; 9: 443-9
Many respiratory interventions performed in the ICU, such as the nebulised administration of drugs, intubation and extubation, high flow nasal O2, and non-invasive ventilation, are considered aerosol generating. 8, 9
We describe a novel personal ventilation hood that provides a physical barrier between the patient and environment, negative pressure within the hood, and high efficiency particulate air (HEPA) filtration of expired air. 11
Aerosol particle measurementsData were obtained from an Aerodynamic Particle Sizer (APS) spectrometer (TSI APS 3220) and a Scanning Mobility Particle Sizer (SMPS) spectrometer (TSI SMPS 3080, with 3081 column). The APS counts aerosol numbers and sizes of aerosol with aerodynamic diameters between 0.5 μm and 20 μm (most of the respiratory size range) 6
Aerosol measurements occurred in a clean room (volume 34 m3, HEPA-filtered with six air changes per hour) at the Department of Mechanical Engineering, University of Melbourne, Australia. Clean rooms are defined by the International Organization for Standardization (ISO: 14644-1: 2015) 16
To our knowledge, no agreed definition of the degree of aerosol particle generation related to medical procedures exists. The ISO clean room standard defines that the maximum number of particles 0.5 μm or greater in an ISO 7 (pharmaceutical) clean room is 0.352 particles per mL. 16
The personal ventilation hoodThe hood (Figure 1) consists of a mobile steel and plastic frame, a fan and a standard HEPA H13 filter (rated to 99.97% clearance of 0.3μm particles). The plastic barrier opens out to a hood of about 1.3 m3 volume. The barrier contains droplets, while the ventilation reduces aerosol escape. Air is drawn from the sides and front near the patient’s legs up to the rear, and routinely away from the health care worker (principally at the front and sides of the patient). The air passes through the fan (Westaflex, Melbourne, Australia; airflow = 40 L/s) and a HEPA H13 filter (Techtronic Industries, Hong Kong, China), thereafter returning to surrounding air.
Volunteer testingEthics approval for this study of droplet and aerosol (aerosol) measurements from one healthy, male volunteer (first author) was deemed not required by the Western Health Research Ethics manager. Only the volunteer entered the clean room; the scientist sat immediately outside monitoring aerosol concentrations. Aerosol concentrations rose upon the volunteer’s room entry. Experiments did not commence until levels returned to baseline; typically after 10–20 minutes. Measurements were performed on 9 and 14–16 April 2020 and in the first week of May 2020.
The aerosol inlet sampled air at 6.8 L/min from outside of where the plastic hood folded down, about at average face height for a health care worker. The inlet was 18 cm rightward of the bed at a floor height of 155 cm, 65 cm directly away from the patient’s head and 80 cm from the hood articulation point (where the plastic hood is pulled down).
We measured aerosolisation in seven scenarios:
- quiet breathing;
- talking (counting loudly continuously);
- wet coughing;
- dry face mask at 15 L/min O2 flow;
- high flow O2 (30 L/min and 60 L/min) humidified nasal cannulae;
- bilevel non-invasive positive-pressure (BiPAP) (5 positive end-expiratory pressure [PEEP], plus 10 continuous positive airway pressure [CPAP]) with humidification; and
- 5 mL nebulised saline with face mask at O2 10 L/min.
Breathing, talking and coughing were directed towards the measurement port (worst case scenario of a patient–health care worker interaction). Wet cough was achieved with 5 seconds of forced coughing, 25 seconds resting, and drinking 25 mL water, then repeated. High flow humidified nasal prongs therapy was administered by an Airvo (F&P 950; Fisher and Paykel, Auckland, New Zealand). Non-invasive ventilation was administered using a face mask by a Dräger Oxylog 3000 (Lübeck, Germany) with humidification with an F&P 950 humidifier. The non-invasive ventilation mask was deliberately set to be “leaky”, to simulate a worst case aerosol generation scenario, by inserting a nasogastric tube 15 cm within the mask.
Repeated nebuliser aerosolisation against the personal ventilation hoodTo test the ventilation hood further, we used compressed air and the same nebuliser mask alone (no volunteer) as an aerosol source placed 50 cm above bed level. We ran two experiments: one with the aerosol generation similar to the nebuliser and one at a similar level to BiPAP; 15 tests planned for each experimental group. We added 5 mL 0.9% saline to the nebuliser with each 15-minute testing run. We performed 15 test pairs (hood down/fan on; hood up/fan off) for each source strength.
Data analysisDescriptive statistics were calculated for the volunteer’s testing and for the 30 planned, repeated nebulisation tests: mean, median, total particle numbers, and particle sizes. The particle size distribution with 25th–75th and 10th-90th percentiles were displayed as box-and-whisker (Tukey) plots. The mean aerosol concentrations from the prior background count were subtracted to determine the perturbations above background for each test.
Background aerosol counts testingWe used a clean room level equivalent to ISO-14644-1: 2015 Level 7: less than 0.2 particles per mL for particles greater than 0.5 μm. 16
Volunteer testingFigure 2 displays the APS results on a logarithmic y-axis from breathing and talking through to nebulisation. Aerosol generation was similar from wet coughing, dry O2 at 15 L/min via face mask, and high flow (30 L/min or 60 L/min) humidified air via nasal prongs. All were substantially less than one aerosol particle per mL. Humidified, leaky BiPAP (10 cmH20 CPAP, 5 cmH20 PEEP) had about a 100-fold greater aerosol generation compared with high flow nasal cannulae. Nebulised 0.9% saline (a source of aerosols) with O2 at 10 L/min increased aerosol generation a further 30-fold beyond BiPAP.
The distribution of aerosol particle sizes is shown in Figure 3. Aerosol counts at all sizes are dominated by nebulisation, followed by non-invasive ventilation. Wet coughing, dry O2 at 15 L/min via face mask, and high flow (30 L/min or 60 L/min) humidified air via nasal prongs all had similar aerosol quantities and size distributions.
The Online Appendix, section 2, presents detailed results for the aerosol counts and diameters for all aerosol generating scenarios:
- breathing quietly;
- wet coughing;
- dry face mask at 15 L/min O2 flow;
- high air flow (30 L/min and 60 L/min) via nasal cannulae;
- leaky (nasogastric tube inserted) non-invasive ventilation (10 CPAP plus 5 PEEP); and
- 10 L/min via nebuliser.
The application of the hood during non-invasive ventilation and nebuliser therapy reduced aerosol counts measured by SMPS and APS at a typical location of a health care worker by over 98% to a low level (Table 1).
From the originally planned sample of 30 pairs of tests, one APS and three SMPS measurements were excluded because experimental conditions did not equilibrate. Efficiencies were calculated as the averages of these 29 (APS) and 27 (SMPS) tests. Figure 5 shows the percentage of hood efficiency (ie, percentage of nebulised aerosols captured by the hood) and associated data. For Experiment 1 (nebuliser levels), the APS/SMPS aerosol counts per mL for inside and outside the ventilation hood were 2.9/1100 and 1100/77 000 respectively. For experiment 2 (nebuliser aerosolisation similar to BiPAP levels), the APS/SMPS counts per mL for inside and outside the ventilation hood were 0.49/120 and 280/12 000 respectively. The efficiencies were calculated individually for each experiment pair with the different source strengths, then averaged together to create the final efficiencies and the size versus efficiency curve. Overall, the hood was shown to remove more than 99.7% of particles larger than 0.5 μm (those most relevant to human generation) and more than 98.1% of those smaller than 0.5 μm.
DiscussionFor the first time, to our knowledge, we evaluated aerosol generation associated with common respiratory therapies in a clean room. We found that non-invasive ventilation and nebuliser therapy were significant aerosol generating procedures. However, high flow, humidified nasal O2 (60 L/min) and O2 via face mask (15 L/min) only caused a slight increase in the aerosol count. A novel personal ventilation hood reduced the very large number of 0.5–5 μm aerosols (most of the normal human range) 6
Several recent studies exist of the effects of high flow nasal cannulae on aerosol spread. Loh and colleagues 17
Most aerosols (< 5 μm diameter) and droplets (> 5 μm diameter) 2
Our pre-clinical study has limitations: we sampled the environment at one position only (albeit where a health care worker is often likely caring for a patient) and tested one healthy volunteer (not a patient). Limitations in our system meant that the sample inlet was not ideal for particle sizes larger than 5 μm. The volunteer’s breathing, talking and coughing aerosol counts and distributions were similar to prior studies. 6, 21
Our human volunteer study provides high quality data results that may have widespread implications regarding the safe use of ubiquitous respiratory therapies. Our findings that O2 therapy by face mask up to 15 L/min and high flow nasal O2 therapy up to 60 L/min only marginally increased particle counts suggest that these procedures are unlikely to lead to clinically significant aerosol generation and support the use of contact and droplet precautions when using these therapies in patients with respiratory infectious diseases. However, non-invasive ventilation and nebulisation are high aerosol generating procedures, supporting the ongoing use of appropriate personal protective equipment for contact and airborne precautions. 9
The personal ventilation hood reduces repeated nebuliser aerosolisation particle counts by 98%. The ventilation hood may reduce cross-contamination of respiratory infectious diseases beyond SARS-CoV-2 (eg, tuberculosis and influenza) more effectively than wearing an N95 mask (N95 respirator). An effective barrier between the patient and the health care worker addresses concerns about our first line of defence against cross-contamination from airborne infectious diseases, protecting the air we breathe. These preliminary effectiveness data support the evaluation of the hood in a clinical environment.
ConclusionWe confirmed that non-invasive ventilation and nebuliser therapy are high aerosol generating procedures, but standard face mask and high flow nasal O2 therapy are not. The use of a personal ventilation hood substantially reduced the increase in aerosol counts observed with both non-invasive ventilation and nebuliser therapy.
Acknowledgements: We thank Jo Staines (Industrial Director), Geoff Duke, Max Rounds and Aditya Ramani from the Department of Mechanical Engineering, University of Melbourne; and Samantha Bates, Miriam Towns, Teresa Squillacioti, Gypsy Stevenson, Alex Angel, Annette Kakris and Marion Kainer from Western Health.